What does the Nova StatStrip meter measure?
The Nova StatStrip Glucose Hospital Meter used with the Nova StatStrip Glucose Test Strips is a single use test system used to quantitatively measure blood glucose levels, also known as blood sugar, from fresh capillary (fingertip) whole blood, venous and arterial and neonate.
What is the function of the stat strip blood glucose meter?
The StatStrip Glucose Hospital Meter System is a point-of-care, handheld blood glucose monitoring system that was specifically designed to automatically correct for interferences and provide accuracy comparable to the central laboratory.
What is the reportable range for the Nova StatStrip meter?
a. The reportable range of the meter is 10-600 mg/dL. Test results will appear in 6 seconds.
What type of battery is in the Nova meter?
Your monitor uses a CR2450 3-volt coin cell battery. To begin using your monitor, you need to install the enclosed battery.
What type of battery is in the StatStrip meter?
Enhance your purchase
|Number of Batteries||1 Lithium ion batteries required. (included)|
|Battery Cell Composition||Lithium Ion|
|Unit Count||1 Count|
What is a Nova Stat Strip?
StatStrip is designed specifically to measure and correct for clinical interferences that can be present in critically ill patients. Proven safety and accuracy. In one study of over 17,300 capillary results from critically ill patients that were compared to lab results.
How do you reset the Nova glucose meter?
If the meter appears to be unresponsive or frozen, reset the meter by removing the battery for a few seconds and then put the battery back on. Make sure that the battery is not drained. Change the battery or place the meter onto the Charging/Docking Station.
What type of blood can be used with the Nova StatStrip meter and the Nova test strips?
4.4. Specimen type: The Nova StatStrip® is designed for use with fresh capillary, arterial, venous, neonatal heel stick or neonatal arterial whole blood. The most common usage is capillary whole blood by finger-stick.
How long are Nova test strips good for?
Once opened, test strips are stable at room temperature for up to 180 days (approximately 6 months) or until the manufacturer’s expiration date, whichever comes first. ALWAYS write the Date Opened and Discard Date on the vial. Do NOT reuse test strips. It should be disposed after a single use.
How long are QC vials good for?
The QC solutions can be used until the expiry date on the vial label or for 90 days from the date they were first opened, whichever comes first.
What is statsensor® creatinine?
StatSensor® Creatinine is a handheld analyser and miniaturised, disposable biosensor for whole blood creatinine testing. StatSensor® incorporates a patented Multi-Well technology adapted from Nova’s hospital glucose monitoring system.
How many biosensors has Nova introduced in the past 30 years?
During the past 30 years, Nova has introduced 20 biosensors, including the industry’s first biosensor to directly measure whole blood glucose in 1988, and the industry’s first biosensor to directly measure whole blood creatinine in 1996.
What is statsensor® point of care?
Nova StatSensor® Point-of-Care Creatinine and eGFR Measuring System StatSensor® Creatinine is a handheld analyser and miniaturised, disposable biosensor for whole blood creatinine testing. StatSensor® incorporates a patented Multi-Well technology adapted from Nova’s hospital glucose monitoring system.
What is statsensor tmcreatinine?
Nova StatSensor TMCreatinine Measuring System In response to the growing need to quickly and easily assess kidney function in many point-of-care settings, Nova Biomedical introduces StatSensor Creatinine, a handheld analyzer and miniaturized, disposable biosensor for whole blood creatinine testing.